Authors:Mhaveer Singh, Yunus Kamal, Ennus Tamboli, Rabea Parveen, Shahid Ansari, and Sayeed Ahmad
A novel HPTLC method has been developed for the estimation of glabridin in Licorice rhizome and its Unani polyherbal formulation (Qurs-e-Gul). Separation was achieved on silica using toluene, dichloromethane, and ethyl acetate in equal ratios. A compact, well resolved peak of glabridin with RF value 0.56 ± 0.02 was observed. Calibration curve revealed a good linear relationship with r2 value of 0.993 between the peak area and concentration in the range of 25–500 ng spot−1. The proposed method was validated as per the International Conference on Harmonization (ICH) guidelines. The stability assessment was carried out by studying degradation of glabridin stressed by acid, base, oxidation, thermal, and humidity. Photodegradation was also carried out after keeping the drug in sunlight, dark, and in UV lights. The method proposed can be used for routine determination of glabridin in crude drugs and in herbal formulations containing Licorice as one of the ingredients, for quality control as well as for stability testing with high precision, accuracy and a wide range of linearity.
Authors:Y.T. Kamal, Mhaveer Singh, Shahana Salam, and Sayeed Ahmad
The alternative system of medicines like Unani and Ayurveda is preferred worldwide nowadays due to its therapeutic efficacy, lower side effects, holistic approach, psychological dimensions, and qualitative action of weather and seasonal requirement. A simple procedure is described for the simultaneous extraction and estimation of piperlongumine and piperine in a well-known Unani polyherbal formulation using reversed-phase high-performance liquid chromatography (HPLC). The chromatography was carried out on reversed-phase C18 (250 × 4.6 mm) column with a mobile phase containing acetonitrile—water (50:50 v/v). Detection was accomplished with ultraviolet (UV) detection at λ = 325 nm. The flow rate was kept as 1.0 mL−1. The proposed method was validated according to International Conference on Harmonization (ICH) guidelines for accuracy (94.4–105.0%), precision (0.37–2.17% RSD), and robustness (0.14–2.11% RSD). The limit of detection (LOD) values were found as 30 and 10 ng mL−1, while limit of quantification (LOQ) was 100 and 30 ng mL−1 for piperlongumine and piperine, respectively, which proved the sensitivity of the method satisfactory enough for accurate analysis of the both piperlongumine and piperine.